|Back to receiver comparison page
Available as special buy
Available on loan
|Serviceable at UNAVCO
File Size (24 hr, 15 s)
|Time lasts recording at 15 s
Power draw w/ antenna
|Time lasts on 18amphr battery
The Trimble NetRS is used throughout the EarthScope Plate Boundary Observatory Network (PBO). The NetRS has been discontinued by Trimble but continues to be supported by UNAVCO. Firmware version 1.3-2 is recommended for all users and can be installed on all NetRS receivers regarless of Warranty Date.
Configure via an http interface; no special programming software.
Download via the http interface or via ftp; no special downloading software.
No ftp push capabilities.
No screen; LED lights indicate operating mode.
Records L1, L2. Records L2C with firmware v1.1-5 and above. For more information regarding L2C go to: Trimble NetRS - How to track and log L2C
Data stored on internal compact flash card.
Native data format is .T00, or .T01 with firmware 1.2 and above; streams RT17, or RT27 with firmware 1.2 and above.
There is no special software needed for the Trimble NetRS.
UNAVCO NetRS configuration file (.cfg)
This file can be uploaded to the NetRS via its web interface. The receiver will be configured with a static IP address of 192.168.1.2 when the configuration file is enabled.
Power Draw Profile for Trimble NetRS and Iridium Systems
UNAVCO performed a benchtop test to record a high-resolution time history of power consumed by a Trimble NetRS GPS system with Iridium communications.
Power Draw Tests
UNAVCO performed a short benchtop test series to determine the difference in power draw by a Trimble NetRS receiver when connected to three common geodetic GPS antennas.
Article ID: 471
Created: February 1, 2013
Last Updated: January 5, 2018
Author: Freddy Blume
Online URL: https://kb.unavco.org/article/trimble-netrs-resource-page-471.html