GNSS Receiver Evaluations 2012
This article provides a summary of UNAVCO's comparison of the current state-of-the art hardware from 5 major manufacturers. It will include comparisons of receiver data quality in the following areas: double difference phase residual RMS; qualitative comparisons of L2C signal to noise observations (S2); and tracking performance.
For the results presented in this article we used the following testing configuration:
Note: If you would like access to any of the data used for this report, please email Henry Berglund at email@example.com.
Phase Residual RMS
The following figure shows the average phase residual RMS from double-differenced carrier-phase processing using the GAMIT software package. The RMS is binned by elevation angle along the x-axis in 5-degree intervals. The y-axis is RMS in millimeters. Each bin is a 5-day average of the computed LC RMS (LC = 2.546*L1 – 1.984*L2). At all elevations the RMS of the Javad SIGMA was the highest. This may indicate that the default carrier tracking loop bandwidth of the Javad Sigma is greater than the other tested receivers. Only three of the tested receivers allow adjustments to the tracking loop bandwidth: Javad Sigma, Topcon Net-G3A and Septentrio PolaRx4 Pro. As tracking loop bandwidth increases, dynamic performance of a given receiver will also improve. Conversely, noise performance will decrease as tracking loop bandwidth increases (RMS). Addtional testing of receiver performance in dynamic environments need to be undertaken before determining the optimal balance of minimizing noise while maintiaining lock during shaking.
L2C Signal to Noise
The following figure shows the L2C signal to noise from PRN12. The x-axis is in hours and the y-axis is in dB-Hz. Each curve is offset by 10 dB-Hz for clarity. From top to bottom we have: the PolaRx4 Pro, the Sigma, the NetR9, the NetRS and the GR10. The SNR resolution of these receivers varies from 0.05-0.25 dB-Hz. The Leica GR10 has the highest output resolution. While trying to plot the SNR data from the Topcon Net-G3A receiver in 0.1 dB-Hz resolution, a firmware bug was discovered and reported to TOPCON. Due to the bug, we were not able to include the results from the Topcon Net-G3A with this report. Future firmware releases should address the issue.
The following figure shows the average RMS of the MP1 and MP2 combinations over a 5-day period for each receiver. These values were computed by Teqc.
The following figure shows the number of recorded observations divided by the number of expected observations for each reciever. These values were computed by teqc and a 10-degree elevation mask was used. All of the receivers logged more than 99% of the expected observations. Only the Javad Sigma and Leica GR10 logged 100% of the expected observations. This is for GPS L1 and L2 signals only.
The following figures show the average number of detected cycle slips that occured during our 5-day test. The upper figure shows the number of slips that were detected for satellites with elevations angles greater than 10 degrees. The lower figure shows the number of slips that were detected for satellites with elevation angles less than 10 degrees.
|Posted by: Henry Berglund - Wed, Jul 11, 2012 at 10:00 PM. This article has been viewed 6524 times.|
|Online URL: https://kb.unavco.org/kb/article/gnss-receiver-evaluations-2012-742.html|