Critical! Trimble NetR9 FW 4.41 and 4.43 Addendum
If you upgrade a Trimble NetR9 receiver from a firmware version number < 4.41 to a firmware version number 4.41 -> 4.43, then your receiver may not reboot automatically. The voltage supplied to the receiver must be > 15 V for the receiver to automatically power up after applying the firmware update. If the voltage supplied to the receiver is < 15 V, then the receiver will not power up automatically and will require an operator to press the power button.
The Trimble NetR8 model is not affected by this behavior.
THIS HAS BEEN FIXED in v4.46 and later versions. Please use the latest version of the firmware to avoid potential problems.
The following notification was placed on trimble.com under the respective v4.41, v4.42 and v4.43 firmware download pages:
Name : NetR9
|Posted by: Henry Berglund - May 18, 2012. This article has been viewed 325 times.|
|Online URL: https://kb.unavco.org/article/critical-trimble-netr9-fw-4-41-and-4-43-addendum-736.html|
Powered by PHPKB (Knowledge Base Software)